The Mini-Beadbeater-24 disrupts 4 to 24 microbial or tissue samples with better than 95 percent efficiency. Cells are disrupted quickly and safely in the sealed system. The apparatus is easy to clean, has a small footprint and is essentially maintenance free. The machine offers advanced electronics for enhanced motor function and variable shaking speeds of 2000-3450 strokes/min.
Cat. No. 112011, Mini-BeadBeater-24, 115 volt, with safety shield and 2 ml vial cassette.
Cat. No. 112011EUR, Mini-Beadbeater-24, 230 volts, with safety shield and 2 ml cassette.
NOTE for European buyers! The Mini-Beadbeater-24 is CE certified.
The Mini-BeadBeater-24 disrupts microbial cells and plant and animal tissue by violently agitating four to twentyfour 2 ml screw-cap microvials containing small glass, ceramic or steel beads and 0.1 to 1 ml disruption buffer. Even resistant samples like yeast, spores or fibrous tissue are completely homogenized in 1-3 minutes. The non-foaming, aerosol-free method preserves enzymes and organelles. In the presence of nucleic acid extraction media such as phenol, Gu-SCN or a commercial kit solution, DNA or RNA is recovered in the highest possible yield. The method is ideal for PAGE, PCR applications, and diagnostics using antibody or oligonucleotide probes. Because the beads and vials are disposable, there is absolutely zero cross-contamination between samples - essential for PCR techniques.
Protocols developed using a different model of the Mini-BeadBeater are transferable with minimal modification. The performance of the Mini-BeadBeater equals or exceeds that of any other type or brand of cell disrupter.
The Mini-Beadbeater-24 can also be used for dry grinding. Here, steel beads are added to hard samples such as hair, bone, teeth, seeds and minerals and are completely powdered in 10-60 seconds of operation. Softer materials such as biological tissue, rubber or plastics can be powdered by pre-freezing the sample to liquid nitrogen temperatures (cryo-grinding). Dry gringing requires using special microvials resistant to breakage. When dry grinding with ceramic or steel beads at room temperature our 'XXTuff' microvials are recommended. Our stainless steel microvials are available for dry grinding with steel beads at cryo-temperatures.
- Power: 115 volts, 60 Hz, 7.5 amps or 230 volts, 50 Hz, 3.7 amps
- 10' D X 18' W X 12' H, 50 lbs
- Shaking pattern: Uses proven, more efficient near horizontal vial orientation
- Capacity: four to twenty four screw-cap microvials (0.5, 1.5, and 2.0 ml) each containing to 400 mg (wet weight) bio-sample
- Shaking speed: Continuously variable (analog slider), 2000-3450 strokes/min
- Throw or stroke displacement: 0.75 inches
- Timer: 0-5 minute digital, with auto reset
- Removable vial-holding cassette
- No imposed motor cool-down-time between each sample run
- The Mini-BeadBeater-24 uses standard screw-cap plastic microvials. Stainless steel microvials or special reinforced polypropylene microvials ( XXTuff vials ) are available for dry- or cryo-milling with steel beads. Eight larger capacity, 7 ml vials can be processed using an accessory vial-holding ring (see Parts and Accessories below) .
BioSpec Products manufactures five models of beadbeater cell disrupter. We were the first to introduce the 'beadbeating' cell disruption method 30 years ago. The Minibeadbeater has been a very successful product and has superseded most traditional methods of cell disruption or homogenization. Almost a dozen other manufacturers now offer similar cell disruption machines, many which are well designed and deliver high performance. Look for shaking speeds of at least 2000 rpm and a throw (or displacement) of the 2 ml microvial of at least 3/4 inches, shaken in a 'top to bottom of the vial' direction.
Unfortunately, some of these machines use a term intended to measure shaking efficiency: meters/second. In practice the term is varied by controlling the speed of shaking. A common question BioSpec receives is, "What is m/sec? I am trying to adapt a published protocol that used another manufacturer's machine at their speed setting of 4.5 m/sec. What does that mean and what speed do I set my MiniBeadbeater?"
Answer: Cell disruption performance settings of meters/second attempt to take into consideration both shaking speed and microtube displacement of their machine and express them in a single term. Perhaps the intent of the m/sec term was to unify the mechanical features of their machine. Take as an example, centrifuges. Centrifuge performance is expressed in g-force, not motor speed. The g-force term permits scientists to reproduce a centrifuge's performance, no matter who designed and manufactured it. Unfortunately, there is no comparable performance term for shaking cell disrupters. The m/sec term does not rest on sound physical concepts that define energy of shaking or efficiency of grinding. It hinders comparisons between experimental cell disruption protocols with other laboratories using different bead mill cell disrupters.
Factors influencing cell disruption performance are numerous and complex and cannot be expressed in a simple mathematical or physical term. It is suggested that the following guide-line be followed when when using bead mills designed for cell disruption: Operate the shaking cell disrupter machine at its maximum speed setting. Specialized applications requiring operation of a bead mill at lower speeds are rare. In other words, if the objective is to disrupt cells, crank up the rpm of the machine to maximum and get the job done. The time of beadbeating will depend on the nature of the biological sample and the type of bead mill shaker used. With current high performance machines, 2-3 minutes gives close to 100% cell disruption. If you are doing PCR work with nucleic acids and can settle for less than 100% cell disruption, shorter periods of beadbeating will suffice.
Besides shaking speed and vial throw distance, other important variables which determine efficiency of cell disruption are bead size, bead composition and bead load in the vial. These three variables must be optimized by a user for each particular sample. Your MBB operating instructions gives you guidelines on how to do that, as does this web site. And, our tech guys are there to help you, too.
You May Also Be Interested In...
Cat. No. 10831, 2 ml Sterile vial and cap, sterile, foam rack, 100 count Cat. No. 10832, 2 ml Non-sterile vial and cap, 500 count Cat. No. 3205, 7 ml Non-sterile vial and cap, 1000 count Cat. No. 330TX, 2 ml 'XXTuff' non-sterile vial and cap, 500 count Cat. No. 522S, 2 ml Sterile vial and cap, 500 count Read more >